Strip chart recorders use a long roll or strip of paper to archive data points over a timed interval. Using data acquired by one or more sensors, strip chart use one or
more pens to plot the data linearly as the chart passes at a uniform speed. The benefits of strip chart recorders includes the ease of operation, the hardcopy it produces, and
Common applications for strip chart recorders include temperature and humidity measurements, flow rates, pH, pressure, as well as a host of other process measurements. Strip chart
recorders are especially well suited for recording continuous processes. Strip chart recorders are available in single or multichannel styles and in various configurations. Many
recorders can also record information in a digital format for download to a computer. Strip chart recorders require the correct charts and pens to work correctly.
Strip Chart Recorder Technology
Generally speaking, strip chart recorders simply render data. The data itself is produced by application specific sensors that attach to the recorder. Sensor types include temperature
(thermocouple, thermistor, RTD), strain gauge or bridge, current, humidity, level, pressure, pH, as well as many others.
Sensors can be either active or passive devices. Passive sensors, like thermocouples, do not require an outside power supply to generate a signal. Active sensors, on the other hand, do
require a power supply to generate a signal. Power to active sensors is often provided by the recorder and is referred to as the excitation source. Signal excitation can either be a
voltage or current output.
As the recorder receives a digital signal, analog signal or sensor input, the chart paper (for chart or strip recorders) is driven past the pen (or pens) at a steady rate by the drive
mechanism. The pen assemblies typically use a galvanometer or potentiometer to drive the marking device.
Galvanometric recorders have a light coil of wire magnetically suspended that is deflected by the sensor current. The pen records those fluctuations, effectively
transcribing the sensor current. Sensitivity, however, is limited with galvanometers, particularly with analog instruments. Galvanomteric recorders typically have slow response, small
bandwidth, low input impedance, and low sensitivity.
Potentiometric recorders compare the input signal with a reference which results in an error signal which is amplified thus driving the pen motor. Potentiometric
recorders typically have a fast response , smaller bandwidth, very high input impedance, and high sensitivity.
Some things to Consider When Purchasing Strip Chart Recorders:
Is a visual record preferred?
Does monitoring need to be continuous or is sampling sufficient?
How long will the device be used to record or log?
Will the device be left unattended?
Will the device be in a hazardous area?
How many sensors will be used?
What level of accuracy is required?
If you have any questions regarding strip chart recorders please don't hesitate to speak with one of our engineers by e-mailing us at email@example.com or calling 1-800-884-4967.