Laurel BOX Enclosure
Wall mount polycarbonate enclosure sealed to NEMA 4X with optional slash-proof cover
Need Help? Call a Calibration Engineer 1-800-884-4967
We're open Mo-Th 8am to 5:30pm. Fr 8am to 5pm ET
- Sealed on all sides to IP65 / NEMA-4X
- Includes 1/8 DIN cutout
- Includes three 0.24" (6.1 mm) cable glands
- For 1/8 DIN instruments up to 5.30" (134 mm) deep
- Impact resistant polycarbonate
- Stainless steel hardware
- Easy to mount
- Available with IPC IP67 / NEMA-4X splash-proof cover to protect 1/8 DIN meters from front
This product is an accessory for the following products:
The Laurel BOX is a wall-mount / bulkhead enclosure which provides environmental protection to IP65 / NEMA-4X for an 1/8 DIN instrument. It comes with a 1/8 DIN (45 x 92 mm) panel and accommodated instruments up to 134.5 mm (5.30") deep including connectors. This includes all 1/8 DIN instruments by Laurel Electronics and most 1/8 DIN instruments by other manufacturers.
The BOX has an instrument compartment that is sealed on all sides. The two clamshell halves of the case are joined by an O-ring. The rear mounting holes are outside of the sealed compartment. Captive stainless steel screws and high-impact polycarbonate provide durability.
The panel cutout can be expanded by the user up to 150 x 71 mm (5.90" x 2.80") or 131 mm x 110 mm (5.16" x 4.33") to accommodate instruments larger than 1/8 DIN. Internal molded standoffs with a height of 6.5 mm (0.25") can be snipped off by the user if required.
Cable feedthroughs are via three sealed cable glands at the bottom of the case. These make a tight seal with round cables from 6.1 to 1.00 mm (0.04" to 0.24") in diameter. Note that these feedthroughs are not compatible with off the shelf cables with molded connectors, such as USB cables.
The Laurel BOX may be ordered with the IPC splashproof cover. The combination of Models BOX and IPC will provide IP65 / NEMA-4X protection to 1/8 DIN instruments regardless of manufacturer, provided that the instrument depth is less than 134.5 mm (5.30").